Tech Staff Meeting Notes July 8, 2008
Current Top Priorities are: Station maintenance
The fish fry will be on Friday August 15.
Chris and Jim Napors completed acquisition of a new vehicle for John. John will drive his old vehicle to Memphis the week of August 11, join us for the fish fry on August 15, and return to NC with his new truck. Chris will need to get it ready (e.g. decals, plates, roof rack, winch etc).
There are four levels of priority.
|Priorities||Wish to try to get these done (or make progress on) before the next meeting (yeah right)|
|Second Tier||Fit this in too if an opportunity arises. May be good to do instead of priorities but check with Mitch. Likely to be a priority in a previous and/or subsequent month|
|As Time Permits||Fit these in when you can but not instead of first or second priorities|
|Other||Stuff we hope to get to and don't want to forget about|
The task list updated February 6, 2008. Note that everyone is still needed to work on essentially all tasks. There is however, a single person who is primarily responsible for each one. Mitch of course, is ultimately responsible for everything.
Station Maintenance, NMSZ
Station maintenance is the top priority.
We were very fortunate to have a decent sized event on July 5th just before the meeting. Data are in /gaia/smeagol/dat2/Working/NM/200807/20080705_065750_MAN When you look at the traces for this quake almost every single station/channel were up and clearly showing the event. Only the stations listed below were down.
We should be able to get BROM, CATM & GOBM up soon.
BROM-ISIS canister in shop
CATM-Broken during last service!
CWPT-Telementry link problem, this is going to take some quality time to fix.
Isis Oxidation problem
Down, sporadic or unusable
TIPT replacement. The permission with the TN Department of Corrections is complete. ETN
CCRT - Needs a rebuild after a bad lightning strike. I have the Panda II box here and am realigning it.
NC Node - Need to figure out how to fix GPS time drop-outs. Greg suggests moving the antenna away from the building. I will go in the spring, the Forest Service doesn't recommend I try to drive up there unless I really need to, apparantly the road is passable but not recommended. Jim suggests making sure the clock is configured correctly.
GFM - Need a crew to go back up and repair storm damage and replace presumably bad pre-amp card.
RBNC - Replace battery(s), and add another solar panel. Same deal as NC Node concerning roads though. I tried to go up last week and even though I had a key, I had to stop at the first tunnel because of ice.
ETT Node - Need to rebuild/repair remote reset. We are still not sure if the problem is at Starr Mtn. or at the EMA office. Need to decide whether it would be better to build something in the shop and take it up or try to repair on site.
GMG - E-W is bad, need to check out. Need battery box solar reg mounting plate and need to fuse the battery.
Routine Maintenance Kudzu, battery check, etc, etc.
Second Tier Priorities
Most of the strongmotion stations can run at 100 sps without any modifications. We will install a router on the few that have problems (RDST will be the test case). We will also migrate the acquisition software from gcf2ew on anssworm to scream/scream2ew on a linux or ms server. The latter will facilitate stations maintenance without interrupting the data stream.
On the CMG5TD's, Mitch found that for GNAR and LNXT we must first set the filing mode to filing and then set to dual. Otherwise trigger messages show up in the logs but the waveforms are not stored. This trick did not work at SFTN, RDST, GILT, or HDBT. Why not? Possibly firmware differences?
Dual mux assembly is ready for testing.
MKTA spare air conditioner is repaired. Needs to be returned to the doghouse.
Need new RF preamps, radio align, etc, etc at MKTA.
PEBM and HENM sometimes have telemetry fades. Should reevaluate comms.
Need to knock down the jungle around the guy anchors at NMAD.
Need to finish the tower work at NMAD.
Need to swap out the old node computers at NHIN, and AZTN.
Chris and James will visit MAIR tomorrow to get comm going there. Need the ISP contact info from Steve. Done--online.
CJTR still needs comm and a service visit for data download, routine maintenance, yada, yada.
Still sometimes lose comm with stations going through LCHS. Not sure why but it seems to come back ok.
Greg's BAB (electrical balancer, or Balun, for long power and data runs) doesn't work as advertized. Chris will get the schematics off Greg's computer so John can verify that its assembled correctly.
C1SC and C2SC - The lantronix boxes are in non air conditioned rooms and routinely (especially C2SC) fail in the summer. I've found that if you pull the mss 100 out, wait for 1/2 hour and put them back in, they work fine until it heats up again. Can I modify a box with a cooling fan? Is there another device that deals with the heat of Charleston in the summer?
WMTN - Should we consider digging it up and doing another pit (even though I swore I wouldn't do that) The existing one is eroding away and I just don't know how long it's going to last. We may have to end up having this one in sediment. This is the one Mitch and Greg helped dig the pit for, it's so steep on the side of that hill, that the top side of the pit is 10 ft. deep and the bottom side is 6 ft.
LRVA - We have permission, can install any time. Installed and transmitting data. East/West channel has low amplitude and needs troubleshooting.
WSNC - (Winespring Bald, NC) - Have Special use permit from Forest Service. Need to determine details of exactly what goes precisely where. John will get some pics and discuss via phone with Jim and Chris.
Fort Payne - John found a good spot with decent odds of a permission at Desoto state park in Alabama. It will have to repeat through a Georgia fire tower and John is investigating the permit there.
The map shows stations with yellow triangles being new ones.
|SMNC||4||done (see note above)|
The recorder disks should be reformatted as a routine maintenance task next time access is available. The constant read/write to the fat file systems lead to fragmentation. Don't ya just love dos?
Lost contact with recorders 1, 3, and 6. Can reset all 7 recorders which implies the equipment enclosure at each node has power and that at least the reset radio and telemetry are working.
The base isolation bearings will be undergoing a routine check following the Bellmont earthquake. We need to keep an eye on when this is being done so that we can take advantage of it for some maintenance. Jim will order a couple spare yagies (they're a little fragile).
This is the point where we hop on the TDOT wireless network. Its also a strong motion station. When the funds come through, this will become an 18-component "station" like Mud Island. The surface station needs to move to where the boreholes will be. Chris met with AR DOT people and figured out where everything is supposed to go. Most likely we'll do this later this summer. Sharam dropped off the hardware for testing.
Mud Island vertical array is operational. Not really, functional is a better word until we can get it online to qc/vet data.
Steve spoke with Dean Kemp at NWS. They are willing to let us piggyback on their cable internet (not .nws.noaa.gov). Dean is looking into the possibility of a static IP. This would get NAIT online. Bill called him Dean and didn't have answers to some questions. He'll call him back armed with the IT fact sheet for the mss100's.
Lenox Agreement. Mitch spoke with Ron Martens, EMF Broadcasting. There is a desire for some sort of MOU and a visit to CERI.
This is the FDL Policy.
Four firewire terminals need to be refurbished and sealed.
Need to make ready for deployment all CMG6TD's and K2's.
As Time Permits
CPCT - Re-work lighting and pest chaser.
SWET - Chris found the freewave enclosure full of water, we need to design a rain shield for all the aluminum Hoffman boxes, they are leaking at the hinged side because the weather-stripping isn't making good contact with the frame. He put some good tape on it. The battery box was in good shape. (wasn't the problem the orientation of the box?).
BHT - I have never heard anything from the FAA about moving the station to their site on Black Mtn. to improve access and get stuff out of a tree. Right now everything is in a new tree and working great, so there is no rush until that tree dies and falls over.
3876 Power The power for the new lighting on the east side of the general parking lot to the west of CERI is coming from 3876. Chris will check it out to assess the added lightning hazard and see if anything can be done about it. Might be hard to do without the wiring diagram. Chris will make one if needed as time permits.
Chris is going to make a diagram for 3890. Good luck son.
Installing new smeagol broke mysql. Sid is helping get the tables
hooked back into mysqld. Al and Sid got this fixed.
Jim has all the broadband sensors defined and is working on additional equipment (e.g. isis) and the connections. Mitch will install Bill's account in mysql and inv so he can start building stations. Mitch needs to sit down with Jim and figure out what we need to have ISTI help us with. Sid will be available after about mid-May to finish up anything we need his help to take care of. Apparently the downloaded docs from guralp for the cmg5td's have the old +-4g contants. Mitch has paper documentation with the correct +-2g constants.
Withers needs to determine current orientation for MPHB so we can pull it and orient correctly.
Need to teach someone (our hazmat dude?) to neutralize etchant and clean the etching tanks. Along those same lines, Greg has a ton of drawings that play only with the "Eagle" software on his computer. Mitch is unaware of a way to convert to a more generic format. Also require a printer that does negative (the one on Greg's serial port does, others do not). Mitch knew this dependence on vendor specific software would bite us one day. That day is soon.
Mitch punted the Pink Palace hardware to Gary. James is interested in helping and will speak with Gary. It needs to go digital because the analog signal is about to be blocked by new construction at Central and Highland.
John needs a spectrum analyzer. On order.
Jim will switch the FCC licenses from Greg's name to Mitch.